Most Active Software Boards: Install The installation package for the driver is distributed as an MSI 2. Is there a better method for getting data over an RS connection? For example, if you use the Getting Started VI and then place an Initialize VI in your program, the additional session handle will be invalid. Also, if it’s a USB serial port, sometimes they get reassigned. Based on what you’re saying it does not appear to be.
|Date Added:||19 November 2005|
|File Size:||17.88 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
This program was developed by another student on this same r3644a and worked. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. More Information For more information about this driver and other instrument drivers and software available from Agilent Technologies visit: In either case, serial port settings Baud rate, data bits, etc.
I am very interesting in reading the KnowledgeBase article mentioned above, however, I receive a message saying that I do not have permission to read it.
NET Framework itself is not required by this driver. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. You can download the parser and its associated SDK from: You agient install the driver by double-clicking on the file. Any e36644a or tips for resolving these issues would be appreciated as I keep searching for them. Don’t know if this will help, but check the serial port settings Message 7 of 9.
Agilent E36xx IVI-COM Driver
Llabview contains “Getting Started” information on using the driver in a variety of programming environments as well as documentation on IVI and driver specific methods and properties. Most Active Hardware Boards: Message 5 of 8.
Everything else was left as default. The driver installer calls an installer for these components if they are not already present on the system. If not found, the installer will abort, warn, or install the required component as appropriate. This doesnt seem to be a program issue, but rather a communication between Labview and the Agilent power supply. Message 6 of 9. Auto-suggest helps you quickly narrow avilent your search results by suggesting possible matches as you type.
Using the Agilent test program I am able to set the output voltage. I added COM3 as a hardware asset for that driver session. The supported model numbers are: The possible reason given for this is: You are running into Dot-Comma-issues. If you post your VI a better analysis can be done.
Agilent | Agilent E36xx IVI-COM Driver
BTW, the above vi can be found at: I get error or 0xBFFE saying that the session handle is not valid. There really is only one method.
This driver supports instruments manufactured by Agilent Technologies. What could be the reason for this and how can it be fixed? Message 2 of 9. Hence, I decide to go with the generic IVI class drivers.
Oddly, I haven’t come across a single example that uses the generic green class VIs yet. This method should allow you to run this afilent.